HP 8664A Signal Generator 100 kHz to 3 GHz, +13 to -139.9dBm. AM, and Standard FM Rates up to 2 MHz and Deviations to 20 MHz Make the 8664A Ideal for High-speed Digital Communications. 0.01 Hz Frequency Resolution <-100 dBc Nonharmonic Spurious to 2 GHz Reduces Error, When Testing Spurious Immunity of Receivers. Ideal for out-of-channel Receiver Testing.
The HP 8664A signal generator is suited for basic receiver testing up to 3.0 GHz. Pulse width and delay can be internally adjusted between 50 ns and 999 ns, eliminating the need for an external pulse generator. Standard FM rates of up to 2 MHz and deviations to 10 MHz are suitable for many applications such as high-speed digital communications.
* <-140 dBc/Hz typical SSB phase noise at 20 kHz offset
* <-100 dBc nonharmonic spurious to 2 GHz reduces error when testing spurious immunity of receivers
* AM, FM, and optional pulse modulation
* 0.01 Hz frequency resolution
Range - 0.1 to 3000 MHz (8664A)
Accuracy - Timebase stability x f
SSB phase noise @ 1 GHz (20 kHz offset) –117 dBc/Hz; Nonharmonics: (>10 kHz offset) <–100 dBc, 187.5 to 2060 MHz
Subharmonics: <–75 dBc, 0.1 to 1500 MHz
<–50 dBc, 3000 to 6000 1 MHz
Residual FM @ 1 GHz: 0.3 to 3 kHz post det. BW: <7.5 Hz rms;
Resolution: 0.1 dB
±1 dB, output =–119.9 dBm, 1 to 1000 MHz
±1.5 dB, output
=–119.9 dBm, 1000 to 3000 MHz
=–119.9 dBm, >3000 MHz, <1 MHz
Reverse power protection: 25 W
, 0.1 to 2060 MHz; 1 W, >2060 MHz
=+7 dBm dc to >10 kHz, >10 MHz
Accuracy: ±(6% of setting + 1%) up to 90% depth
Maximum peak deviation (Deviation halves per lower octave)
20 MHz, 3000 to 6000 MHz; 10 MHz, 1500 to 3000 MHz
Bandwidth (3 dB): dc to 800 kHz
Carrier accuracy in FM: ±0.6% of setting
Indicator accuracy: ±9%, <20 kHz rates
Distortion: <1%, 20 Hz to 20 kHz rates
Waveforms and rates Sine; white Gaussian noise (0.1 Hz to 400 kHz); Triangle, sawtooth, square (0.1 Hz to 50 kHz)
Frequency accuracy Same as timebase
Digital sweep Digitally stepped sweep over entire frequency range. Linear/log selection. 0.5 to 1000 s sweeps.
Markers/Z-axis output 3 markers available/Z-axis output nominally + 5 V/X-axis output nominally 0 to 10 V
Phase continuous sweep 40 MHz of span available at maximum carrier frequency. 20 ms to 10 s sweep times
Interface GPIB (IEEE 488.2-1987)
Control language Hewlett-Packard Systems Language (HP-SL). All functions controlled except power
IEEE-488 functions SH1, AH1, T6, TE0, L4, LE0, SR1, RL1, PP0, DC1, DT0, C0, E2
Power requirements ±10% of 100 V, 120 V, 220 V, or 240 V; 48 to 440 Hz; 500 VA
Operation temperature 0°C to 55°C
Leakage Conducted and radiated interference meets MIL-STD-461B RE02 and FTZ 1046
Calibration interval Recommended 3 years (MTBC)
Size 177 mm H x 426 mm W x 624 mm D (7 in x 16.8 in x 24.6 in).
All Items sell with 5 days right of return, and 15 days GUARANTEE on all parts and labor.
If in order to comply with our guarantee, we need to refund your money, you will get a FULL refund for the price of the unit...
If you need to make use of the 15 days Guarantee - We will fix, or replace, your unit. In case every attempt to fix or replace it fails, we’ll refund your money.